In today’s market, it is imperative to be knowledgeable and have an edge over the competition. ACI members have it…they are engaged, informed, and stay up to date by taking advantage of benefits that ACI membership provides them.
Read more about membership
Become an ACI Member
Founded in 1904 and headquartered in Farmington Hills, Michigan, USA, the American Concrete Institute is a leading authority and resource worldwide for the development, dissemination, and adoption of its consensus-based standards, technical resources, educational programs, and proven expertise for individuals and organizations involved in concrete design, construction, and materials, who share a commitment to pursuing the best use of concrete.
American Concrete Institute
38800 Country Club Dr.
Farmington Hills, MI
Feedback via Email
Home > Publications > International Concrete Abstracts Portal
The International Concrete Abstracts Portal is an ACI led collaboration with leading technical organizations from within the international concrete industry and offers the most comprehensive collection of published concrete abstracts.
Title: Calibration of Optical Fiber Refractive Index Sensor with Potential for Structural Corrosion Monitoring
Author(s): C.C.C. Lam, M. Rajesh, T. Sun, K.T.V. Grattan, S.K.T. Grattan, S.E. Taylor, and P.A.M. Basheer
Publication: Special Publication
Appears on pages(s): 73-84
Keywords: chloride; fiber optic sensor; interferometer; long period grating; structural health monitoring
Abstract:There is a need in industry for new devices for the monitoring of chloride ion ingress in structural concrete. This work reports on the development of a reflective, gold-coated long period grating-based senor for the measurement of chloride ions in solution, with potential for evaluating the corrosion condition of concrete structures. The sensor scheme is based around a long period fibre grating (LPG)-based Michelson interferometer where the sensor was calibrated and evaluated in the laboratory using sodium chloride solutions, over a wide range of concentrations, from 0.01 M to 4.00 M. The grating response yields shifts in the spectral characteristic of the interferometer, due to the change of refracting index of the solution surrounding it. It was found that the sensitivity of the device could be enhanced over that obtained from a bare fibre by coating the LPG-based interferometer with gold nanoparticles.
Click here to become an online Journal subscriber