In today’s market, it is imperative to be knowledgeable and have an edge over the competition. ACI members have it…they are engaged, informed, and stay up to date by taking advantage of benefits that ACI membership provides them.
Read more about membership
Become an ACI Member
Founded in 1904 and headquartered in Farmington Hills, Michigan, USA, the American Concrete Institute is a leading authority and resource worldwide for the development, dissemination, and adoption of its consensus-based standards, technical resources, educational programs, and proven expertise for individuals and organizations involved in concrete design, construction, and materials, who share a commitment to pursuing the best use of concrete.
American Concrete Institute
38800 Country Club Dr.
Farmington Hills, MI
Chat with Us Online Now
Feedback via Email
Home > Publications > International Concrete Abstracts Portal
The International Concrete Abstracts Portal is an ACI led collaboration with leading technical organizations from within the international concrete industry and offers the most comprehensive collection of published concrete abstracts.
Title: Grip-Specimen Interaction in Uniaxial Restrained Test
Author(s): S. A. Altoubat and D. A. Lange
Publication: Special Publication
Appears on pages(s): 189-204
Keywords: end effects, restrained shrinkage, shrinkage, tensile creep, tensile test
Abstract:Restrained tests are used to evaluate the risk of early age cracking and the craking sensitivity of concrete mixtures. One test that has become common in recent years is the active uniaxial restrained test in which the length change due to shrinkage is recovered by applying external load to maintain the concrete sample at constant length. The length change is measured by linear variable differential transformer (LVDT), which is used as the control signal in this test. In such tests, the dog-bone geometry is used to grip the ends. To ensure a fully restrained test, the LVDT response to the loads and to shrinkage specimen interaction should not interfere with the measurement of deformation, and this depends on the instrumentation and how the LVDT is attached to the concrete specimen. Some experiments in the literature have the LVDT attached to the steel grips, a practice vulnerable to possible error due to the interaction between the grip and the concrete. This study considered two methods of attaching the LVDT. First, the LVDT is attached to the steel grips, second, the LVDT is attached to the concrete within the zone of reduced cross-section. The results indicate that attaching the LVDT to the grips results in errant measurement of the shrinkage stress, creep, and elastic strains due to the grip-specimen interaction. The consequences will be false interpretation of fully restrained shrinkage and creep characteristics because the grip-specimen interaction leads to a partially restrained test. The study suggests mounting the LVDT to the concrete sample away from the grips to achieve a fully restrained test. Results for two concrete mixtures with w/c ratio of 0.51 and 0.56 are discussed for both methods of attaching LVDTs.
Click here to become an online Journal subscriber